Shenzhen Jiwei Automations Ltd.
Address：5th Floor, Baoli Building, No. 3162 Longgang Avenue, Shenzhen, P. R. China.
Tel：+86 755 28407683
Jiwei Launched NAMUR Output Tuning Fork Level Switch
Similar to 4～20mA two-wire output, NAMUR output is also a two-wire system. But for instrument with NAMUR output, the current supplied is smaller and the power consumption is lower. The instrument with NAMUR output and the one with 4～20mA two-wire output can both be applied in Intrinsic Safety explosion protection area, and AC or DC power supply for both of them is wired through safety barrier.
Just for safety considerations, NAMUR output tuning fork liquid level switch can be replaced for the one with 4～20mA two-wire output. But for instrument with 4～20mA two-wire output, the signal output is in 10～36V and 8mA/16mA, while for the one with NAMUR output, the signal output voltage is in 8.2V，the output current can be equal or lower than 1.2mA and equal or higher than 2.1mA. Because the instrument with NAMUR output works in very small current supplied and its power consumption is very low, it is much more complicated to design and meet its interference resistance than the one with 4～20mA output. Therefore, considering from the instrument configuration aspect or the safety barrier used, the tuning fork liquid lever switch with NAMUR output is more expensive than the one with 4～20mA two-wire output.
NAMUR output type of Tuning Fork Liquid Level Switch Jiwei recently developed and launched can fully compete with those imported from Europe in its safety, stability and interference resistance. This new product is our latest achievement based on our innovation and customer-centric philosophy.
The other news
- 2019-12-05>Jiwei Launched NAMUR Output Tuning Fork Level Switch
- 2019-10-10>Jiwei Magnetic Level Indicator Applied in Level Measurement of Hydrofluoric Acid Medium
- 2019-09-17>Ring-11 Liquid Level Switch Widely Applied in Pump Protection System
- 2019-08-16>Jiwei Level Measuring Instrument Already Obtained 16 Explosion-proof Certificates
- 2019-07-12>Fork-11 Level Switch Successfully Applied in the Measurement of Fumed Silica
- 2019-07-02>Well-known Japanese Flowmeter Enterprise Visits Jiwei to Negotiate Cooperation
- 2019-05-08>Jiwei’s New Design —— An 400℃ Ultra-high Temperature Vibrating Rod Level Switch
- 2019-04-26>Jiwei Successfully Solves the Problem of Liquid Chlorine Measurement